arXiv:1703.07765v1 [cs.ET] 22 Mar 2017

Cooperative In-Vivo Nano-Network Communication at Terahertz Frequencies

Qammer H. Abbasi, Senior Member, IEEE, Ali Arshad Nasir, Member, IEEE, Ke Yang, Member, IEEE, Khalid Qaraqe, Senior Member, IEEE, Akram Alomainy, Senior Member, IEEE

Abstract—Nano devices have great potential to play a vital role in future medical diagnostics and treatment technologies because of its non-invasive nature and ability to reach delicate body sites easily as compared to conventional devices. In this paper, a novel concept of cooperative communication for invivo nano network is presented to enhance the communication among these devices. The effect on the system outage probability performance is conducted for various parameters including relay placement, number of relays, transmit power, bandwidth and carrier frequency. Results show approximately a 10-fold increase in the system outage performance whenever an additional relay is included in the cooperative network, hence show a great potential of using cooperative communication to enhance the performance of nano-network at terahertz frequencies.

Index Terms—nano communication, Terahertz, body area network, channel modeling, cooperative communication.

I. INTRODUCTION

Nano-technology has a critical role nowadays in multidisciplinary domains including biomedical, industrial control, military and environment. However its impact in biomedical domain due to its non invasive nature is making huge impact and driving research in this direction more intensively. Nano-network idea emerged from the connectivity of nano devices, which was followed by nano-communication proposal to enhance the features of these devices [1]. Communication between nano-devices can be performed by different mechanism e.g., molecular, electromagnetic (EM), nanomechanical or acoustic [2] etc. However, EM based communication is considered as a feasible technique in the terahertz band for exchange of data among these nano-machines [3] because biological tissues are non-ionized at these frequencies and also THz band is less susceptibility to scattering phenomenon [4].

During past few years studies on body-centric communication have been gradually increasing [5], [6], however reduction in size requirement is making nano-technologies an attractive choice for body-centric communication. Due to evolution of novel materials like graphene, [7] capable of

This publication was made possible by NPRP grant # 7-125-2-061 from the Qatar National Research Fund (a member of Qatar Foundation). The statements made herein are solely the responsibility of the authors.

Qammer and Khalid are with the Dep. of Electrical and Computer Engineering, Texas A & M University at Qatar. Qammer is also with Queen Mary University of London and UET, Lahore.; e-mail: {qammer.abbasi;k.qaraqe}@tamu.edu.

Ali Arshad Nasir is King Fahd University of Petroleum and Minerals; email:anasir@kfupm.edu.sa

Ke Yang and A. Alomainy are with Antennas & Electromagnetics School of Electronic Engineering and Computer Science Queen Mary University of London, London; e-mail: {ke.yang;a.alomainy}@qmul.ac.uk.

operating at THz frequencies, the interest in these frequencies for communication of nano-devices either on- or in-side the human body is growing as well. In addition, due to molecular resonance at these frequencies, even fine variation in water content or tissues can be detected, hence making this frequency propagation mechanism an emerging area of research for biomedical applications [8]. There are numerous studies presented in literature discussing about the applicability of THz communication in biomedical domain [9]–[13]. A detailed review on the current state-of-the-art technologies and applicability of nano communication in biomedical application is presented by Abbasi et al. in [9], while discussing about physical and networking layer concepts. The applicability of THz in nano-network is studied in [12], [13] and MAC-layer protocol is presented in [14]. Berry et al. [10] characterised the optical properties of human tissues upto 2.5 THz. In order to further enhance the human tissue parameters at THz band, terhertz time domain spectroscopy is used to characterise the skin sample obtained from Blizard Institute in [15]. In this paper, collagen is created with different fibroblast cell numbers to see its effect on biological tissue parameters and signal propagation through the tissue. Ke et al. [16] developed a tissue model and performed numerical and analytical studies inside the body to study pathloss and its relation with the dielectric loss. In [17] (authors of this paper), a novel channel model inside the body at THz frequencies for communication between these nano-devices based on extensive numerical and measurement studies have been presented, while considering various parameters like distance, number of sweat ducts and frequency. All of the above studies showed that, the communication range is very small at these frequencies because of high pathloss and in order to transmit reliable signal to receiver for getting useful information, a new paradigm of communication is needed. Although, as mentioned before there are some studies in open literature with regards to nanocommunication, optical parameters characterisation, channel modeling and applicability of THz communication in the biomedical domain, but there is no published research as per authors’ knowledge about the performance evaluation and advantage of using cooperative communication among these nano-devices.

In this paper, we are presenting for the first time in literature about the applicability and the study on the performance evaluation of cooperative in-vivo nano network communication at terahertz frequencies. Outage probability performance evaluation is performed for various scenarios including different relay placement, i.e., relays placed vertically and randomly;

Figure 1. Envisioned cooperative architecture for in-vivo nano-network

different distances between the transmitter and the receiver, where relays were placed vertically and variation in source power, bandwidth and carrier frequency.

The rest of the paper is organized as follows. Section II presents the system model and simulation setup used in this study. Section III discusses the results for relay based communication considering various scenarios as mentioned above. Finally, conclusions are drawn in Section IV.

II. IN-VIVO NANO-NODES BASED SYSTEM MODEL Body-centric nano-network can be categorised in three domains like traditional body-centric communication i.e., onbody, off-body and in-body. An envisioned architecture [18] for in-vivo nano-network can be summarised as shown in Fig. 1. In Fig. 1, nano-nodes are the simplest and smallest nanodevices to perform simple computation and detection and then transmit it to relay nodes, which are also a small nanodevices and their task is to do amplification and forwarding the received signal to nano-routers. These nano-devices (nanorouters) are slightly larger in terms of computational and behaviour capabilites and can also act as a control unit for the set of nano-nodes by ordering simple commands like read, sleep, wakeup etc. They can be invasive or non-invasive depending on the application. The nano-micro interface is composed of hybrid devices which are used to exchange information between the two interfaces and the last unit i.e., gateway allows the user to remotely control the system using internet.

As mentioned before in the introduction section, since pathloss at terahertz frequencies is considerably large even at very small distance and in order to transmit information to relatively larger distance in this paper multiple relays are used to assist communication between the transmitter nanonodes (Tx) and the nano-router/receiver node (Rx) as shown in Fig. 2 (Note that though Fig. 2 shows the vertical midway placement of relays, however, in Section III, we also analysed the performance of cooperative system with random relay placement). To ensure implementation simplicity at the relays, an amplify-and-forward (AF) relaying assumption has been made.

Figure 2. System model for in-vivo cooperative communication at terahertz frequencies

Due to high pathloss, the direct link between the transmitter and the receiver would result in negligible received signal-

Figure 3. Outage probability performance for varying bandwidth, BW = {0.5, 1, 1.5, 2, 2.5} GHz. Figure 4. Outage probability performance for varying carrier frequency f = {0.5, 0.75, 1, 1.25, 1.5} THz.

to-noise ratio (SNR) at the receiver side and thus it can be ignored. In this paper maximum ratio combining (MRC) has been employed at receiver side i.e., at nano-router [19]. Thus, the received SNR at the Rx, γ_{Rx}, through relaying link is given by [20]:

γ_{Rx} = M∑ i=1 γ_{s,ri} γ_{ri,d} γ_{s,ri} + γ_{ri,d} + 1 , (1)

where γ_{s,ri} ≜^{|hs,ri |2Ps}_{ℓs,ri σ2} , is the received SNR at the i_{th} relay due to the link between transmitter and the i_{th} relay, γ_{ri,d} ≜^{|hri,d|2Pr} ℓ_{ri,d}σ^2 is the received SNR at the destination

node due to the link between the i_{th} relay and the Rx, ℓ_{s,ri} and ℓ_{ri,d} are the path losses from source transmitter to the i_{th} relay and from the i_{th} relay to the destination or Rx, respectively, h_{s,ri} ∼ CN (0, 1) and h_{ri,d} ∼ CN (0, 1) are complex normally distributed channel co-efficients for the respective links, respectively, σ^2 is the variance of additive white Gaussian noise, P_s is the transmit power of source (transmitter) and P_{ri} is the transmit power of the ith relay. The path loss ℓ_{s,ri} or ℓ_{ri,d} at THz frequency by Abbasi et al. as presented in [17] is used in this work and is given by:

ℓ_{s,ri} = −0.2N + 3.98 + (0.44N + 98.48)d^{0.65}_{s,ri} +(0.068N + 2.4)f^{4.07} (2) ℓ_{ri,d} = −0.2N + 3.98 + (0.44N + 98.48)d^{0.65}_{ri,d} +(0.068N + 2.4)f^{4.07}, (3)

where N = 5 is the number of sweat ducts, f is frequency and d_{s,ri} and d_{ri ,d} are the Euclidean distances from source transmitter to the i_{th} relay and from the i_{th} relay to the destination or Rx, respectively.

Performance Metric: Outage probability is chosen as the performance metric to evaluate the communication performance of the considered system, where outage probability in this paper is defined as the probability that received SNR, γ_{Rx} < γ_{th}, where γ_{th} is the threshold SNR for decoding the received signal.

III. DISCUSSION OF RESULTS

In this work communication system for M = {1, . . . , 5 relays is considered. The relays transmit with 100 nW power, i.e., P_{ri} = −40 dBm (100 nW), ∀ i ∈ M. Noise variance per unit bandwidth (BW) is = −174 dBm/Hz and threshold SNR is set to γ_{th} = 10 dB. Unless otherwise specified, the system bandwidth is set to be 1 GHz, carrier frequency to be f = 1 THz, the distance between the transmitter and the Rx to be d = 0.2 mm, and source transmit power is set to P_s = −40 dBm (100 nW). Moreover, unless specified in particular, all relays are placed vertically midway between the transmitter and the receiver, such that the two adjacent relays are 30^∘ angle apart from each other.

Figs. 3 and 4 demonstrate the outage probability performance for different values of bandwidth, BW = {0.5, 1, 1.5, 2, 2.5} GHz and carrier frequency, f = {0.5, 0.75, 1, 1.25, 1.5} THz. It can be clearly observed that the outage probability performance is improved by increasing the number of relays. Fig. 3 shows that outage probability worsens by increasing the BW due to the increase in the noise variance, which decreases the received SNR γ_{Rx}. Similarly, an increase in the carrier frequency increases the path loss (as shown in (2)), which decreases the received SNR γ_{Rx}. A poor outage probability performance for a single relay case is observed as shown in Figs. 3 and 4. However, it can be obviously improved by increasing the source and relay transmit powers from 100 nW.

Fig. 5 shows the outage probability performance for different placement of relays, such that, relays are placed vertically at a horizontal distance of {0.02, 0.06, . . . , 0.18} mm from the transmitter. Fig. 5 shows that mid-way placement of the relays

Figure 5. Outage probability performance for different placement of relays, i.e., relays placed vertically at a horizontal distance of {0.02, 0.06, . . . , 0.18} mm from the transmitter.

Figure 6. Outage probability performance for different distances between transmitter and the receiver, i.e., d = {0.1, 0.2, 0.3, 0.4} mm, where relays are placed vertically mid-way between transmitter and the receiver.

Figure 7. Outage probability performance for varying source transmit power, P_s = {−50, −45, . . . , −20} dBm. Figure 8. Outage probability performance for varying source power, P_s = {−50, −45, . . . , −20} dBm, while also considering random placement of relays within a circle of radius^d_2 mm.

results in the minimum outage, whereas moving relays closer to the source or the Rx increases the outage probability. This is due to the increase in the path loss either for source-to-relays link or for relays-to-Rx link. Fig. 6 shows the outage probability performance for different distances between the transmitter and the receiver, i.e., d = {0.1, 0.2, 0.3, 0.4} mm, where relays are placed vertically mid-way between transmitter and the receiver. As expected, smaller the distance d, minimum is the outage due to minimum path loss. Fig. 7 shows the outage probability performance for varying source transmit power, P_s = {−50, −45, . . . , −20} dBm (10nW, . . . , 10μW ). The outage probability decreases with the increase in the source transmit power. Fig. 7 also shows that further increase in the source transmit power beyond −30 dBm (1μW ) results in marginal improvement in the outage probability. In Fig. 8, same results with some additional curves (dashed lines) are presented, which consider random placement of relays within a circle of radius^d_2 mm. Fig. 8 shows that random placement of relays worsens the outage probability performance, which is also supported by the findings of Fig. 5.

IV. CONCLUSION In this paper, studies on cooperative communication for nano-scale electromagnetic based communication inside the body is presented. Various scenarios for relay based communication is considered including relay location, variation in source power, bandwidth and carrier frequency variations etc. Results highlight the increase in system performance by 10- fold when using cooperative communication hence paving a new direction of research for applicability of single or multiple relay based communication in nano network for improving system performance.

REFERENCES

[1] I. F. Akyildiz, F. Brunetti, and C. Blázquez, “Nanonetworks: A new communication paradigm,” Computer Networks, vol. 52, no. 12, pp. 2260–2279, 2008.

[2] A. M. Andrew, “Nanomedicine, volume 1: Basic capabilities,” Kybernetes, vol. 29, no. 9/10, pp. 1333–1340, 2000.

[3] S. Balasubramaniam and J. Kangasharju, “Realizing the internet of nano things: Challenges, solutions, and applications,” Computer, vol. 46, no. 2, pp. 62–68, Feb 2013.

[4] G. Piro, K. Yang, G. Boggia, N. Chopra, L. Grieco, and A. Alomainy, “Terahertz communications in human tissues at the nano-scale for healthcare applications,” Nanotechnology, IEEE Transactions on, 2015.

[5] Q. H. Abbasi, M. U. Rehman, K. Qaraqe, and A. Alomainy, Advances in Body-Centric Wireless Communications: Applications and State-of-the-art. The Institution of Engineering and Technolog (IET) Publication, July, 2016.

[6] Q. H. Abbasi, A. Sani, A. Alomainy, and Y. Hao, “Numerical characterization and modeling of subject-specific ultrawideband body-centric radio channels and systems for healthcare applications,” Information Technology in Biomedicine, IEEE Transactions on, vol. 16, no. 2, pp. 221–227, 2012.

[7] S. Luryi, J. Xu, and A. Zaslavsky, Future Trends in Microelectronics:

Frontiers and Innovations. John Wiley & Sons, 2013.

[8] T. Binzoni, A. Vogel, A. H. Gandjbakhche, and R. Marchesini, “Detection limits of multi-spectral optical imaging under the skin surface,” Physics in medicine and biology, vol. 53, pp. 617–636, 2008.

[9] Q. H. Abbasi, K. Yang, N. Chopra, J. M. Jornet, N. AbuAli, K. Qaraqe, and A. Alomainy, “Nano-communication for biomedical applications: A review on the state-of-the-art from physical layers to novel networking concepts,” IEEE Access Journal, vol. 4, pp. 3920 – 3935, August, 2016.

[10] E. Berry, A. J. Fitzgerald, N. N. Zinov’ev, G. C. Walker, S. Homer- Vanniasinkam, C. D. Sudworth, R. E. Miles, J. M. Chamberlain, and M. A. Smith, “Optical properties of tissue measured using terahertzpulsed imaging,” in Medical Imaging 2003. International Society for Optics and Photonics, 2003, pp. 459–470.

[11] A. Fitzgerald, E. Berry, N. Zinov’ev, S. Homer-Vanniasinkam, R. Miles, J. Chamberlain, and M. Smith, “Catalogue of human tissue optical properties at terahertz frequencies,” Journal of Biological Physics, vol. 29, no. 2-3, pp. 123–128, 2003.

[12] J. M. Jornet and I. F. Akyildiz, “Channel modeling and capacity analysis of electromagnetic wireless nanonetworks in the terahertz band,” IEEE Transactions on Wireless Communications, vol. 10, no. 10, pp. 3211– 3221, Oct. 2011.

[13] I. F. Akyildiz and J. M. Jornet, “Electromagnetic wireless nanosensor networks,” Nano Communication Networks (Elsevier) Journal, vol. 1, no. 1, pp. 3–19, Mar. 2010.

[14] J. M. Jornet, J. Capdevila-Pujol, and J. Sole-Pareta, “Phlame: A physical layer aware mac protocol for electromagnetic nanonetworks in the terahertz band,” Nano Communication Networks (Elsevier) Journal, vol. 3, no. 1, pp. 74 – 81, 2012.

[15] N. Chopr, K. Yang, Q. H. Abbasi, K. Qaraqe, M. Philpott, and A. Alomainy, “Thz time domain spectroscopy of human skin tissue for in-body nano-networks,” IEEE Transactions on THz Science and Technology, pp. 1 – 7, August, 2016.

[16] k. Yang, A. Pellegrini, M. O. Munoz, A. Brizzi, A. Alomainy, and Y. Hao, “Numerical analysis and characterization of THz propagation channel for body-centric nano-communications,” IEEE Transactions on Terahertz Science and technology, vol. 5, pp. 419–426, May 2015.

[17] Q. H. Abbasi, H. E. Sallabi, N. Chopra, K. Yang, K. Qaraqe, and A. Alomainy, “Terahertz channel characterisation inside the human skin at the nano-scale,” IEEE Transactions on THz Science and Technology, vol. 6, no. 3, pp. 427 – 434, May, 2016.

[18] I. F. Akyildiz and J. M. Jornet, “The internet of nano-things,” IEEE Wireless Communications Magazine, vol. 17, no. 6, pp. 58–63, Dec. 2010.

[19] A. Goldsmith, Wireless Communications. Cambridge University Press, 2005.

[20] S. S. Soliman and N. C. Beaulieu, “Exact analysis of dual-hop AF maximum end-to-end SNR relay selection,” IEEE Transactions on Communications, vol. 60, no. 8, pp. 2135–2145, August 2012.