arXiv:1502.07878v1 [physics.ins-det] 27 Feb 2015

Characterization, 1064 nm photon signals and background events of a tungsten TES detector for the ALPS experiment

J. Dreyling-Eschweiler^{a∗∗}, N. Bastidon^b, B. Döbrich^a, D. Horns^b, F. Januschek^a , and A. Lindner^a.

aDeutsches Elektronen-Synchrotron (DESY), Hamburg, Germany; bInstitut für Experimentalphysik, Universität Hamburg, Germany

February 27, 2015 (DESY 14-193) Accepted by Journal of Modern Optics (DOI: 10.1080/09500340.2015.1021723)

Abstract

The high efficiency, low-background, and single-photon detection with transition-edge sensors (TES) is making this type of detector attractive in widely different types of application. In this paper, we present first characterizations of a TES to be used in the Any Light Particle Search (ALPS) experiment searching for new fundamental ultra-light particles. Firstly, we describe the setup and the main components of the ALPS TES detector (TES, millikelvin-cryostat and SQUID read-out) and their performances. Secondly, we explain a dedicated analysis method for singlephoton spectroscopy and rejection of non-photon background. Finally, we report on results from extensive background measurements. Considering an event-selection, optimized for a wavelength of 1064 nm, we achieved a background suppression of ∼ 10^{−3} with a ∼ 50 % efficiency for photons passing the selection. The resulting overall efficiency was 23 % with a dark count rate of 8.6 · 10^{−3} s^{−1}. We observed that pile-up events of thermal photons are the main background component.

∗∗Corresponding author. Email: jan.dreyling-eschweiler@desy.de

Contents

1 Introduction 2

2 Components 3 2.1 Sensor: Fiber-coupled NIST TES . . . . . . . . . . . . . . . . . . 3 2.2 Millikelvin-temperature: Adiabatic Demagnetization Refrigerator (ADR) . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 4 2.3 Read-out: Low-noise SQUID . . . . . . . . . . . . . . . . . . . . 6

3 Performance tests 6 3.1 Stability . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 6 3.2 Linearity . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 7

4 Event analysis and signal region 8 4.1 Pulse shape analysis (PSA) . . . . . . . . . . . . . . . . . . . . . 8 4.2 Signal calibration and event classes . . . . . . . . . . . . . . . . . 9

5 Background events and dark count rate 10 5.1 Intrinsic background . . . . . . . . . . . . . . . . . . . . . . . . . 11 5.2 Thermal photonic background . . . . . . . . . . . . . . . . . . . . 12

6 Summary 14

1 Introduction

Over the last few years, the development of optical transition-edge sensors used as single-photon detectors has resulted in a near-unity detection efficiency [1, 2]. Besides the application in Quantum Optics experiments [3, e.g.], there is an application in the Any Light Particle Search (ALPS) experiment, which we describe in this paper.

The ALPS experiment is a light-shining-through-a-wall experiment [4, 5]: Photons are stored in front of an optical barrier, behind which, a photon detector is placed. Standard Model physics does not allow for light passing the wall with a measurable rate. However, photons could convert into low-mass particles beyond the Standard Model, which would interact very weakly with the barrier. In this way they could effectively traverse the wall and reconvert into a photon behind the obstacle. Such new particles are weakly interacting sub-eV particles (WISPs) [6], including the axion which has been proposed to solve the strong CP-problem [7, 8, 9], are well motivated by theoretical considerations, and could explain several astronomical and cosmological observations [10, 11, e.g.] including non-baryonic Dark Matter [12, e.g].

Using an enhanced setup including an increased magnetic length, optical cavities to amplify the signal and a low-background single-photon detector, the sensitivity of the second generation of ALPS [5] will be improved by three orders of magnitude with respect to the previous generation of this type of experiment [4]. In order to reach the sensitivity required, the single-photon detector should have an overall detection efficiency > 70 % for photons with a wavelength of

1064 nm (and an energy of 1.165 eV), while registering a background not exceeding a few mHz [5]. We found that a transition-edge sensor (TES) is the most promising detection technique [13].

A TES acts as a sensitive microcalorimeter operated in the superconducting transition [14]. Here, small temperature changes, like the absorption of an infrared photon (∼ 1 eV), cause measurable changes of the electrical resistance. Due to its calorimetric nature – events with higher energy result in a higher resistance change – a TES acts as a non-dispersive single-photon spectrometer with a coarse energy resolution. In the range from optical to near-infrared photons, it is about ΔE/E ≲ 10 % [1, 2].

In this paper, we describe the ALPS TES detector and its main components (Sec. 2). We report on results of stability and linearity measurements (Sec. 3). We explain a newly developed method to analyze the pulse shape of events and to calibrate the TES (Sec. 4). We report on our results of long-term background measurements and of resulting dark count rates (Sec. 5). Finally, we summarize our characterization of the ALPS TES detector (Sec. 6).

2 Components

The first efforts to install and operate a TES detector for the ALPS experiment commenced in 2011 [15]. Since the end of 2013, a completed TES detector system has been operated routinely in the ALPS laboratory, namely the ALPS TES detector (Fig. 1). We briefly explain the three main components and their performance. A detailed description is found in [16].

2.1 Sensor: Fiber-coupled NIST TES

We are using a TES provided by NIST^1 [17]. The sensitive material is a 20 nm thin tungsten film which has a sensitive area of 25 μm × 25 μm. The film is incorporated in an optical stack enhancing the detection efficiency of 1064 nm photons. The TES chip is surrounded by a standard fiber connector sleeve (Fig. 1(b)) which allows to connect a single mode fiber [18]. This method is cryogenically compatible and is estimated to show a low optical loss of < 0.1 % in between the single mode fiber diameter (< 10 μm) and the TES area. For such fiber-coupled TES, a detection efficiency of higher than 95 % was reached [17]. Knowing that this is possible in principle, our main aim in these first studies was to investigate backgrounds relevant for the ALPS experiment. For the measurements presented in this paper, we have used standard SMF28 fibers with an anti-reflective coating for 1064 nm.^2

In the commissioning phase of the detector system, we determined further important parameters of the used TES chip^3 [16]. We have determined a superconducting transition of T_c = 140 mK with a width of ΔT_c = 1-2 mK. All following results are referring to the same chosen working point:

• a bath temperature of T_b = 80 mK and

1National Institute of Standards and Technology.

2The coating reduces the standard 4 %-loss at a fiber end to < 1 %. 3In the current setup, we operate two TES chips from NIST: channel A and B. In this paper, we only refer to channel A which is characterized in more detail. For channel B, we have found comparable results.

• a TES set point, R_0, within its superconducting transition of 30 % of the normal resistance, R_N = 13.8 Ω, which is reached by applying a constant bias voltage across the TES.

Thus, the TES has been operated in a strong negative electro-thermal feedback (ETF): Due to the constant bias voltage, the Joule power P_J = V^2/R at the TES corresponds to the power flow to the bath. If the TES heats up, the resistance increases, and the Joule heating is decreased. Thus, the cooling of the TES is faster. Furthermore, the ETF stabilizes the operation as well as simplifies theoretical analysis of the TES [14].

The heat capacity, C = 0.6 fJ/K, and the thermal conductance, G = 15 pW/K, are calculated according to experience with similar optical tungstenbased TES [19]. This results in a thermal time constant τ = C/G ≈ 38 μs. Due to the ETF, the time constant of a TES is effectively reduced to τ_{eff} =^τ_{1+α/n} , where α = T/R · ∂R/∂T which is the steepness of the resistance-temperature dependence at the set point R_0 and n expresses the thermal impedance between the bath substrate and the superconducting film which is 5 or 6 for tungsten [20]. Using an averaged pulse, we determine τ_{eff} = 1.53 μs, such that α/n = 24 and therefore find α to be between 120 (n = 5) and 144 (n = 6). Furthermore, we determined an energy loss of ∼ 10 % when comparing the measured pulse to the nominal photon energy [14]. This is assumed to be explained by high-energetic phonons escaping from the tungsten film during the absorption process, so that only a fraction of the photon energy is measured, as described in [21]. We also estimated the TES saturation – when the TES becomes temporarily normal resistive due to an energy input – which would correspond to a 330 nm single-photon event or of more than three simultaneous 1064 nm single photons. The effect of saturation becomes already evident in the linearity measurement for 405 nm photons (Sec. 3.2 and Fig. 2(b)). Finally, a measured average photon pulse has been proven to be in accordance with the small-signal TES theory [14] by using the determined TES parameters.

2.2 Millikelvin-temperature: Adiabatic Demagnetization Refrigerator (ADR)

To reach mK-temperatures, we operate a cryogen-free cryostat, namely an Adiabatic Demagnetization Refrigerator (ADR). The ADR system from Entropy GmbH is constructed following a top-down principle where mK-temperatures are reached at the innermost stage (Figs. 1(a) and 1(c)). A pulse tube cooler with a closed^4He gas cycle provides the precooling below 4 K. The adiabatic demagnetization cooling [22] is realized by a salt pills unit acting as a paramagnetic spin-system, which is surrounded by a superconducting magnet. When the field of the magnet is ramped up to 6 T, the spin-system is aligned. After the pills unit is thermalized, it is thermally disconnected from the precooling stage. By ramping down the magnetic field, a low-temperature below 100 mK can be reached at the innermost stage.

The evacuation and precooling take ∼ 25 h depending on the installed cold mass [16]. Charging or recharging the pills unit takes ∼ 90 min depending mainly on the performance of the adjustable heat switch driven by a piezo motor. Regulating a small remnant magnetic field allows to stabilize the inner temperature to 80 mK ± 25 μK (rms) for ≳ 20 h. A recharge with a subsequent

(a) sketch of ALPS TES detector

(b) sensor module with two channels

(c) opened ADR cryostat

Figure 1: Sketch and components of the ALPS TES detector: (a) shows a sketch of the ALPS TES detector incorporating the different cooling stages of the ADR, the TES device and the optical fiber which is guided outside of the cryostat. (b) shows a sensor module with two channels. The round NIST TES chips can be found on the left surrounded by a fiber sleeve. On the right, the PTB SQUID chip incorporating two channels is located. Aluminum wire bonds are used for electrical connection. (c) shows the opened ADR cryostat with the different cooling stages. The pulse-tube cooler cools the upper copper plate. On this plate, the salt pills unit with the magnet is mounted. The superconducting devices are located on the top, inside the copper mK-shield. In the back, a part of the 20 m long helium pipes connecting the PT cooler to a compressor can be seen.

mK-regulation is named recharge-cycle in the following. We could operate the ALPS TES detector for instance in a 24 h cycle consisting of 20 h measurement time and 4 h dead time due to the recharge and sensor adjustments.

2.3 Read-out: Low-noise SQUID

The TES chip has been integrated in an electrical circuit for two reasons. Firstly, this allows to bias accurately the device in its superconducting transition. Secondly, it enables measurements of the TES resistance variations. Therefore, the TES circuit is inductively coupled to a SQUID^4 sensor. We are using a 2-stage SQUID (Fig. 1(b)) from PTB^5 [23] which provides a low-noise current read-out of the TES circuit. The SQUID device is stably working at mK-temperatures and has a noise level of approximately 2 pA/ √ Hz. The TES and SQUID chips are mounted to the inner ADR stage (Figs. 1(a) and 1(b)). Superconducting or screened cables are used to connect the sensors to the electronics located outside the ADR cryostat. Using an electronics from Magnicon GmbH, we can set bias currents for the TES and SQUID as well as read out the signal output. The read-out system has a bandwidth of f_{3dB} ≈ 0.9 MHz. To record the signal output, we have used a commercial oscilloscope^6. We set the sampling interval to Δt = 50 ns and the digitization step to 0.4 mV. For signal calibration (Sec. 4.2), we have recorded time series with 20 Msamples covering each a 1 s window. For long-term background measurements (Sec. 5), we have recorded triggered single events with an extended read-out window going from 5 μs before to 15 μs after the trigger. We observe a dead time of the oscilloscope below 0.5 seconds when recording a single event. Concerning the long-term background measurements (Sec. 5), we adjust the trigger level to have a trigger rate below 0.1 s^{−1} and not to be affected by this dead time.

3 Performance tests

3.1 Stability

The stability of the ALPS TES detector and thus its reliability is of special importance for the ALPS experiment, where long-term measurements are planned, looking for possible signal photons having a very low rate. Thus, the signal region needs to be stably defined without a necessity to readjust it in short time intervals.

We have investigated the stability of the TES output operated in the ADR with regards to

• different times during one recharge-cycle, i.e. after recharging the pills unit when regulating with a remnant magnetic field,

• different recharge-cycles,

• different cool-downs (after allowing the system to thermalize to room temperature and cooling it down again) and

4Superconducting Quantum Interference Device. 5Physikalisch-Technische Bundesanstalt. 6Tektronix DPO7104C^®.

(a) stability measurement (b) linearity measurement

Figure 2: (a) Stability: the TES bias current equivalent to R_0 = 30 % R_N as a function of time after the beginning of a recharge. (b) Linearity: average pulse height in units of voltage output as a function of photon energy for the TES. The error bars show the widths of the respective Gaussian distributions. The dashed line is a fit to the first three points.

• human operator due to the adjustment method.

With regards to all these factors the system was found to be reasonably stable, i.e. the working point and the signal region do not differ considerably.

Of special importance for the ALPS experiment is the stability within a recharge-cycle to reduce the overall uncertainties. We were able to show that our working point, i.e. the TES bias current for the chosen R_0 (Sec. 2.1), is approximately stable during the course of a recharge. The values of the bias current only vary by about < 1.5 μA in maximum (Fig. 2(a)) which results in a < 3 %-change of the TES pulse height.

3.2 Linearity

We have carried out linearity measurements regarding the response of our TES detector setup to photons of different wavelengths. For that purpose, we used lasers of four different wavelengths in the optical and near-infrared.^7 For each wavelength, we set the laser rate with attenuators to reduce the event rate such that the pile-up rate was negligible. We recorded several pulse height distributions. All the laser sources resulted in approximately Gaussian distributions of the pulse peak height. The different signal regions were fitted with a Gaussian. The absolute widths of the Gaussians were approximately constant due to the constant sensor noise, while it results in relative widths of 4 % (at 405 nm) to 10 % (at 1064 nm) of the corresponding mean values. By plotting the mean values of the Gaussians as a function of photon energies (Fig. 2(b)), it can be seen that the TES response is indeed approximately linear in the region relevant to the ALPS experiment, though results indicate a non-linear behavior for higher energies, which can be understood as a saturation of the detector (Sec. 2.1).

7Nominal wavelength of 405 nm, 532 nm, 635 nm and 1064 nm.

4 Event analysis and signal region

An event is selected if the registered amplitude crosses the chosen (online or offline) trigger level (T L). The corresponding event window includes data points before and after the trigger point, t = 0 (Sec. 2.3). These data points are analyzed to extract different parameters. By using such parameters, events can be classified.

The first common approach is to consider the pulse height (P H) as the maximum level of an event window. This allows to accumulate events in a pulse height distribution (PHD) and to define a one-dimensional signal region. A second parameter is calculated by taking into account the pulse integral (P I). Combining the two parameters in a two-dimensional distribution, it is possible to discriminate between single-photon events with different energies as well as, pile-up events or baseline shifts causing a trigger [16]. Finally, the available information of the pulse-shape can be extracted by fitting an averaged pulse profile to the read-out window. This pulse shape analysis (PSA) achieves the best background discrimination.

4.1 Pulse shape analysis (PSA)

In this analysis, an expected single-photon pulse shape is fitted to an event. As the expected pulse shape, we use an averaged pulse of single 1064 nm photon signals.^8 The average pulse is shifted by an integer multiple j of the sampling interval Δt (Sec. 2.3) before or after the trigger point. For each shift we calculate the best fit by scaling the average pulse shape with a dimensionless pulse scaling factor a.

Thus, the fit function f_j is

f_j (t, a) = a · avg_j (t) ,

where avg_j is the average pulse shifted by the index j. The best fit is determined by choosing a such that χ^2_j is minimized. It is defined as

χ^2_j = N∑ i=1 ( y_i − f_j (t_i, a) dy_i )_2 . We consider N data points of each event window and of the average pulse. y_i is the i-th voltage entry of the event, and f_j (t_i, a) the i-th voltage entry of the scaled average pulse. We determine the uncertainty dy_i by considering the standard deviation of the noise band σ_{noise}. We fix dy_i = σ_{noise} = 5.0 ± 0.1 mV for each data point.

Finally, we can determine the best shift j by finding the minimal χ^2_j . Thus, by analyzing an event with the PSA, we extract two useful quantities: the min(χ^2_j ) ≡ χ^2 and the corresponding scaling factor a of the average pulse. A typical event with the best fit is illustrated in Fig. 3(a).

8The averaged pulse results from measurements including only events which are lying in the 3σ-region of the extended two-dimensional PHD. Using the average pulse as the expected photon pulse shape, the limited read-out bandwidth of the read-out system is taken into account automatically. Furthermore, the expected pulse is a self-consistent estimate of the overall system and does not depend on the first-order TES theory [14].

(a) single event analysis (b) 1064 nm signal region

Figure 3: Application of the pulse shape analysis (PSA): (a) shows the best fit to a single event. The thick gray line in the upper plot illustrates the average 1064 nm pulse. The grayish area illustrates the range of shifting the index, here j = −50 . . . 50. The lower plot illustrates the significant minimum of chi-squared due to the best fit. (b) shows the twodimensional distribution of signal 1064 nm photons. This allows to define elliptic σ-regions. Using 3σ-cuts along both axes, it is found that the signal sample contains few impurities.

4.2 Signal calibration and event classes

To calibrate the ALPS TES detector, we operate the TES with a fiber attached to a single-photon source. As a source we use an attenuated, continuous wave, 1064 nm laser.^9 The attenuation is adjusted to reach a single-photon rate of 10^3 to 10^4 s^{−1}.

We analyze 5 seconds of time series offline. Firstly, we extract single events by applying a trigger level which is 3σ below the mean pulse height in a onedimensional PHD. This excludes low-energetic events with smaller values of pulse heights. Secondly, we apply the PSA method to each extracted event. In the following, we classify the events of the 5 s sample by considering their distribution in the χ^2_{red.}-a-plane (Fig. 3(b)):^{10}

• Signal events (events in the 3σ-region): It is assumed that these events are single signal photons from the laser source. This is validated by analyzing a sample with laser off (e.g. Fig. 5). Signal events normally distribute around μ_a = 1.0 and μ_χ2_{red.} = 0.95. By fitting a two-dimensional Gaussian distribution, we define 1-, 2- and 3σ signal regions which are illustrated by the elliptic contours.

• Low-energetic single photon-like events (χ^2_{red.} < μ_χ2_{red.} + 3σ_χ2_{red.} = 1.34 and a < μ_a − 3σ_a = 0.82): Few events of this sample are presum-

9The nominal wavelength of the used laser is 1063.9 nm. The corresponding energy deviation is much smaller than the energy resolution of the TES detector, ΔE/E < 10 %.

10Instead of the χ^2, we consider the reduced chi-squared which is defined by χ^2_{red.} =^{χ2}_ν , where ν is the degree of freedom. In this example, it is ν = N − n = 400 − 2 = 398, which includes all N = 400 data points of an event window and n = 2 due to two free fit parameters, the scaling factor a and the shift index j.

Table 1: Energy calibration of the fit parameter a: The calibration results from 1064 nm single photons. The standard deviation σ and the corresponding regions result from the PSA of the representative signal sample. a E [eV] λ [nm] mean μ 1.01 1.165 1064 (full) 3σ-region 0.82-1.20 0.948-1.382 1308-897 half (high-energetic) 3σ-region 1.01-1.20 1.165-1.382 1064-897

ably related to low-energetic photons between ∼1300-1950 nm. If these events are due to single photons, they presumably originate from thermal background or from outlying signal events. For this sample, we expect approximately 8 signal events to be located in this region to be compared to the 12 events detected.

• High-energetic single photon-like events (χ^2_{red.} < μ_χ2_{red.} + 3σ_χ2_{red.} = 1.34 and a > μ_a +3σ_a = 1.20): Few events are presumably related to single high-energetic photons between ∼600-900 nm. If these events are due to single photons, they could originate from stray light or from outlying signal events. For this sample, we expect approximately 8 signal events to be located in this region to be compared to the 18 events detected.

• Pile-up or background events (χ^2_{red.} > μ_χ2_{red.} + 3σ_χ2_{red.} = 1.34): For large values of χ^2_{red.}, the contribution of outlying signal events is small (8 expected out of 29 detected events). Most likely these additional events are first-order pile-up events. A first order photon-like pile-up event is defined as two photon-like events that are detected within the same time window. It is shown that such events are well discriminated by the PSA if the two constituents occur within a time window ≳ 0.5 μs (Sec. 5.2). Other background events can also result in a high χ^2_{red.} (Sec. 5.1).

Furthermore, we can determine an energy resolution for this sample related to 1064 nm photons:

(ΔE/E)_{1064 nm} = |σ_a/μ_a| ≈ 6.2 % . (1)

Finally, the scaling factor a provides a calibration. a = 1 is calibrated to the pulse height of the average pulse. Thus, a directly determines the energy of a well fitted event. The calibration for photon energies and corresponding wavelengths is listed in Table 1.

5 Background events and dark count rate

Because an ALPS detector is required to have low dark count rates (Sec. 1), we extensively studied the background of the ALPS TES detector. Using different fiber setups, we undertook several long-term measurements (at least > 15 h), and analyzed the trigger events by applying PSA. Dark counts are defined as events which lie in the signal region. A dark count rate, DC, is calculated by dividing the total counts by the total measurement time. In the following, we mainly present two measurements and analyses:

• Using no fiber and a dark TES in order to analyze the intrinsic background (Sec. 5.1).

• Using a fiber-coupled TES in order to analyze the thermal photonic background resulting from the warm fiber end (Sec. 5.2).

5.1 Intrinsic background

Firstly, we investigated possible noise events of the read-out system [16]. Therefore, we operated only the SQUID with a normal resistive TES. We observed few events with a short-time baseline shift. We interpret these events as environmental electromagnetic interference. For example, switching on and off the laboratory lights or maybe starting a machine adjacent to the laboratory can cause such baseline shifts. All these events are removed by PSA and do not result in dark counts, however, they may lead to an increase of the dead time fraction of < 0.02 %, defined as the ratio of total accumulated dead time and measurement time.

Secondly, we analyzed the intrinsic background of the ALPS detector [16]. Therefore, we operated a “dark” TES chip without a fiber, and the chip surface was facing the 80 mK shield. We determined three populations of calorimetric pulses recorded with this setup (Fig. 4). Besides single photon-like events, we have observed two background populations with a greater value of the exponential falling time constant:

• single photon-like events with τ_{eff} = 1.53 μs (Sec. 2.1) and a rate of ∼ 1 · 10^{−4} s^{−1} (see below)

• 1^{st} background population with τ_1 ≈ 13.1 μs and a rate of ∼ 5 · 10^{−3} s^{−1}

• 2^{nd} background population with τ_2 ≈ 73.0 μs and a rate of ∼ 5 · 10^{−3} s^{−1}

The origin of these two background populations is most likely related to radioactivity and cosmic muons. Firstly, any charged particle could ionize silicon layers of the TES chips and cause a heat up. The two populations could be related to two different types of silicon layers around the tungsten film. This could explain two different thermal time constants. Secondly, both populations show a uniform broad energy spectrum. Thirdly, an estimate of the event rate fits roughly the measured rate. Fourthly, we observed several coincidence events when operating both TES channels, which are located close together. Such background events are discriminated by the PSA method and do not contribute to a dark count rate. The dead time contribution of these background events is < 0.001 % and thus negligible.

The photon-like events could be photons occurring during an ionization process of the sensor chip. A second possible explanation are thermal photons from warmer regions inside the cryostat. We observe that the thermal shields inside the ADR are not totally light-tight. In addition, the spectrum of photon-like events is not uniform and shows more low-energetic events, which could point to a black body spectrum (Sec. 5.2).

Applying the PSA to four long-term measurements, we determined 22 dark counts [16], which lie within the 3σ-region of 1064 nm photon signals (Tab. 1). This results in an intrinsic dark count rate of

DC_{intrinsic} = (1.0 ± 0.2) · 10^{−4} s^{−1}.

Figure 4: Intrinsic background: The plot shows the normalized and averaged pulse of different populations: a single-photon like population and two background populations with longer time constants.

We consider this value as a fundamental limit of dark noise when detecting 1064 nm signals with this TES detector setup. Further reduction is possible after improving the light-tightness of the inner ADR and optimizing the design to avoid photons resulting from an ionization of the chip substrate.

5.2 Thermal photonic background

The dark count rate of a fiber-coupled TES, DC_{fiber}, is necessary for an application. Here, the “warm” fiber end is located outside the cryostat in order to connect it to an optical experiment. It has been observed by other groups that photons from room-temperature surfaces represent a significant background [21]. The spectrum of thermal photons is in accordance with a room-temperature (300 K) black body spectrum folded by the fiber transmittance. For example, there is a cutoff for wavelengths > 2000 nm for silica fibers. However, other groups concentrated mainly on the telecommunication wavelength of 1550 nm as a signal [24, 25, 26]. A dark count rate for 1064 nm signals has to our knowledge not been published.

To determine DC_{fiber} for 1064 nm signals, we performed several long-term measurements [16]. The total fiber was 2.2 m long and its end, outside of the cryostat, was covered to suppress stray light. Due to the limited readout bandwidth, we had to adjust the trigger level to reduce the trigger rate, otherwise measurements would have been affected by dead time due to a high rate of low-energetic thermal photons. We set the trigger to a level which corresponds to a value of nearly a = 1. In the following, we consider the highenergetic half of the 3σ-region (Tab. 1).

Applying the PSA to two representative long-term measurements, we deter-

Figure 5: Thermal photonic background: It is shown a typical result of a long-term measurement analyzed with PSA. We only consider the high-energetic half of the 3σ-region due to the adjusted trigger level.

mined that [16]:

DC_{fiber} = (8.6 ± 1.1) · 10^{−3} s^{−1} . (2)

This value corresponds to the rate for photons with a wavelength going from 897 to 1064 nm including the detector’s energy resolution (Eqn. 1). Furthermore, it corresponds to an overall detection efficiency of ∼ 23 % for 1064 nm signals. This includes optical losses, the chosen trigger level and the efficiency of the analysis.^{11}

In Fig. 5, the distribution of events, the signal region and the corresponding trigger level are depicted. Two photon-like populations are identified:

• “photon-baseline noise” pile-up events: This population is located around a = 0.7 with χ^2_{red.} < 1.5. These events are related to low-energetic photons which can reach the trigger level because of a coincidental noise contribution near the peak height.

• “photon-photon” pile-up events: This population has about a = 1.0 with a range of χ^2_{red.} = 0.7 . . . 3.0. By considering an event with χ^2_{red.} > μ_χ2_{red.} + 3σ_χ2_{red.} = 1.34, it clearly appears as a first-order pile-up event of two photonic constituents. By considering events with χ^2_{red.} ≈ μ_χ2_{red.} + 3σ_χ2_{red.} = 1.34, we can estimate a time resolution of the PSA method, τ_{PSA} ≈ 0.5 μs. If two photon-like events occur within τ_{PSA}, this firstorder pile-up event cannot be discriminated by the PSA and is counted

11 The optical losses are mainly caused by the used fiber setup. Using a commercial power meter, we measured a transmission of only ∼ 60 % due to two fibers with a fiber connector in between. In addition, we conservatively estimate a 95 %-efficiency of the fiber-to-TES coupling (Sec. 2.1). Because of the chosen trigger level, only ∼ 80 % of the nominal signals are considered. Furthermore, the two-dimensional Gaussian 3σ-region contains 98.9 % nominally. Finally, the half 3σ-region includes a factor 0.5.

as a single photon-like event and, in case of matching energy, as a dark count of 1064 nm signals.

It seems that the half 3σ-region is dominated by “photon-photon” pile-up events (Fig. 5). This assumption is discussed in the following paragraph. We compare the measured rate to an expected rate using

1. a model with a black body spectrum and a detector with a perfect time resolution (idealized),

2. a measured thermal photon rate with a limited time resolution (realistic).

First, for an idealized model, we assume a 300 K black body spectrum, a single-photon detector with 10 % energy resolution, perfect time resolution, no optical losses and overestimated parameters for the fiber core area and aperture. This results in DC_{ideal. model} = 3.4 · 10^{−4} s^{−1} [16]. Thus, the measured rate (Eqn. 2) is ∼ 1.5 orders of magnitude higher than DC_{ideal. model}. This suggests that the measured dark count rate cannot be dominated by single thermal photons.

Second, we estimate an effective rate, ṅ_{eff} , caused by first-order pile-ups by considering accidental coincidences. It is [27]

ṅ_{eff} = 2 τ ṅ_1 ṅ_2 , (3)

where τ is the resolving time and ṅ_1 and ṅ_2 uncorrelated rates. We set τ = τ_{PSA} ≈ 0.5 μs as current time resolution of the ALPS TES detector. We estimate ṅ_1 ≈ ṅ_2 ≈ 10^2 s^{−1} from a pulse height measurement (Fig. 4.3b in [16]). Here, ṅ_1 contains thermal photons with wavelengths going from 1600 to 1900 nm and ṅ_2 from approximately 1900 to 2200 nm. Photons of these two contributions can effectively combine to single photon-like events lying in the half 3σ-region of 1064 nm signals. It is ṅ_{eff} ≃ 10^{−2} s^{−1}, which is in the same order of magnitude as the observed dark count rate in Eqn. 2.

6 Summary

The NIST-fabricated TES has been successfully operated in an ADR. Routine operations in the ALPS lab environment have been carried out with measurements on the performance of the system including stability, characteristics of the TES sensor and its readout as well as background measurements. A pulseshape analysis has been developed using a self-consistent pulse template to fit individual pulses. Energy reconstruction and background rejection have been demonstrated to work well (relative energy resolution of ΔE/E ≈ 6 % and background rejection removing spurious events from ionization as well as electromagnetic induced events).

The overall detection efficiency of ∼ 23 % with a background rate of ∼ 9 mHz (Eqn. 2) from a warm fiber has been achieved with this first setup. The main requirements of the ALPS setup (single-photon detection efficiency > 70 % and background rate in the mHz regime) are achievable with an improved setup. The reduction of thermal background is possible with a fiber cutting off at a smaller wavelength. This will relax the required event selection to achieve an efficiency as required for the ALPS experiment.

Acknowledgements

We want to thank NIST, Boulder, US, for the TES devices and PTB, Berlin, Germany, for the SQUID sensors. We also want to thank our ALPS collaborators, especially S. Ghazaryan, E. von Seggern and C. Weinsheimer.

References

[1] Adriana E. Lita, Aaron J. Miller, and Sae Woo Nam. Counting nearinfrared single-photons with 95% efficiency. Opt. Express, 16(5):3032–3040, Mar 2008.

[2] Daiji Fukuda, Go Fujii, Takayuki Numata, Kuniaki Amemiya, Akio Yoshizawa, et al. Titanium-based transition-edge photon number resolving detector with 98% detection efficiency with index-matched small-gap fiber coupling. Opt. Express, 19(2):870–875, Jan 2011.

[3] Marissa Giustina, Alexandra Mech, Sven Ramelow, Bernhard Wittmann, Johannes Kofler, Jorn Beyer, et al. Bell violation using entangled photons without the fair-sampling assumption. Nature, 497(7448):227–230, May 2013.

[4] Klaus Ehret, Maik Frede, Samvel Ghazaryan, Matthias Hildebrandt, Ernst- Axel Knabbe, et al. New alps results on hidden-sector lightweights. Physics Letters B, 689(4-5):149 – 155, 2010. arXiv:hep-ex/1004.1313.

[5] R. Bähre, B. Döbrich, J. Dreyling-Eschweiler, S. Ghazaryan, R. Hodajerdi, D. Horns, et al. Any light particle search ii – technical design report. Journal of Instrumentation, 8(09):T09001, 2013. arXiv:physics/1302.5647.

[6] Joerg Jaeckel and Andreas Ringwald. The Low-Energy Frontier of Particle Physics. Ann.Rev.Nucl.Part.Sci., 60:405–437, 2010. arXiv:hepph/1002.0329.

[7] R.D. Peccei and Helen R. Quinn. CP conservation in the presence of instantons. Phys.Rev.Lett., 38:1440–1443, 1977.

[8] Frank Wilczek. Problem of strong P and T invariance in the presence of instantons. Phys.Rev.Lett., 40:279–282, 1978.

[9] Steven Weinberg. A new light boson? Phys.Rev.Lett., 40:223–226, 1978.

[10] J.L. Hewett, H. Weerts, R. Brock, J.N. Butler, B.C.K. Casey, et al. Fundamental physics at the intensity frontier. 2012. arXiv:hep-ex/1205.2671.

[11] Rouven Essig, John A. Jaros, William Wester, P. Hansson Adrian, S. Andreas, et al. Dark sectors and new, light, weakly-coupled particles. 2013. arXiv:hep-ph/1311.0029.

[12] Paola Arias, Davide Cadamuro, Mark Goodsell, Joerg Jaeckel, Javier Redondo, et al. WISPy cold dark matter. JCAP, 1206:013, 2012. arXiv:hepph/1201.5902.

[13] Jan Eike von Seggern. Overview of low-flux detectors. Proceedings of the 9th Patras Workshop on Axions, WIMPs and WISPs, in Mainz (Germany), 2013. arXiv:physics/1310.0660.

[14] K. D. Irwin and G. C. Hilton. Transition-edge sensors. Cryogenic Particle Detection, pages 63–149, 2005.

[15] Jan Dreyling-Eschweiler, Dieter Horns, Friederike Januschek, and Axel for the ALPS-II collaboration Lindner. Detecting an infrared photon within an hour – transition-edge detector at alps-ii. Proceedings of the 9th Patras Workshop on Axions, WIMPs and WISPs, in Mainz (Germany), 2013. arXiv:physics/1309.5024.

[16] Jan Dreyling-Eschweiler. A superconducting microcalorimeter for low-flux detection of near-infrared single photons. Ph.D. thesis, 2014.

[17] A. E. Lita, B. Calkins, L. A. Pellouchoud, A. J. Miller, and S. Nam. Superconducting transition-edge sensors optimized for high-efficiency photonnumber resolving detectors. Society of Photo-Optical Instrumentation Engineers (SPIE) Conference Series, 7681, April 2010.

[18] Aaron J. Miller, Adriana E. Lita, Brice Calkins, Igor Vayshenker, Steven M.

Gruber, and Sae Woo Nam. Compact cryogenic self-aligning fiber-to-detector coupling with losses below one percent. Opt. Express, 19(10):9102– 9110, May 2011.

[19] A.E. Lita, D. Rosenberg, S.W. Nam, A.J. Miller, D. Balzar, L.M. Kaatz, and R.E. Schwall. Tuning of tungsten thin film superconducting transition temperature for fabrication of photon number resolving detectors. IEEE Transactions on Applied Superconductiviy, 25, Jun 2005.

[20] K.D. Irwin. An application of electrothermal feedback for high resolution cryogenic particle detection. Applied Physics Letters, 66:1998–2000, 1995.

[21] B. Cabrera, R. M. Clarke, P. Colling, A. J. Miller, S. Nam, and R. W.

Romani. Detection of single infrared, optical, and ultraviolet photons using superconducting transition edge sensors. Applied Physics Letters, 73:735, August 1998.

[22] Guy K. White and Philip J. Meeson. Experimental techniques in lowtemperature physics. 2002.

[23] D. Drung, C. Assmann, J. Beyer, A. Kirste, M. Peters, F. Ruede, and T. Schurig. Highly sensitive and easy-to-use squid sensors. IEEE Transactions on Applied Superconductivity, 17:699–704, jun 2007.

[24] Danna Rosenberg, Adriana E. Lita, Aaron J. Miller, and Sae Woo Nam.

Noise-free high-efficiency photon-number-resolving detectors. Phys. Rev. A, 71:061803, Jun 2005.

[25] A.J. Miller, A. Lita, D. Rosenberg, S. Gruber, and S.W. Nam. Superconducting photon number resolving detectors: performance and promise. Proceedings of the 8th International Conference on Quantum Communication, Measurement and Computing, pages 445–450, 2007.

[26] Go Fujii, Daiji Fukuda, Takayuki Numata, Akio Yoshizawa, Hidemi Tsuchida, and Shuichiro Inoue. Effect of black body derived photon emission to superconducting optical detectors. Poster at LTD14, Heidelberg, 2011.

[27] Carl Eckart and Francis R. Shonka. Accidental coincidences in counter circuits. Phys. Rev., 53:752–756, May 1938.