arXiv:1308.6713v1 [physics.optics] 30 Aug 2013

Novel techniques in VUV high-resolution spectroscopy

W. Ubachs^1, E. J. Salumbides^1, K. S. E. Eikema^1, N. de Oliveira^2, L. Nahon^2

1 Department of Physics and Astronomy, LaserLaB, VU University, De Boelelaan 1081, 1081 HV Amsterdam, The Netherlands 2Synchrotron Soleil, Ormes des Merisiers St Aubin BP 45, 91192 Gif sur Yvette Cedex, France

Abstract

Novel VUV sources and techniques for VUV spectroscopy are reviewed. Laserbased VUV sources have been developed via non-linear upconversion of laser pulses in the nanosecond (ns), the picosecond (ps), and femtosecond (fs) domain, and are applied in high-resolution gas phase spectroscopic studies. While the ns and ps pulsed laser sources, at Fourier-transform limited bandwidths, are used in wavelength scanning spectroscopy, the fs laser source is used in a two-pulse time delayed mode. In addition a Fourier-transform spectrometer for high resolution gas-phase spectroscopic studies in the VUV is described, exhibiting the multiplex advantage to measure many resonances simultaneously.

1 Introduction

In the wavelength region of the vacuum ultraviolet, i.e. the range 50−200 nm, most molecules exhibit an unstructured continuum spectrum, since the high energy photons probe the level structure above one or more dissociation limits. However, some important molecules exhibit a spectrum of narrow lines even at wavelengths as short as 100 nm. The carbon monoxide (CO) molecule has its first dissociation limit beyond 10 eV, and its spectrum in the range 80-120 nm is characterized by strongly perturbed rovibronic resonances, where virtually all lines undergo predissociation. These phenomena are of great importance for the investigation of the chemical dynamics of the interstellar medium,

Email address: w.m.g.ubachs@vu.nl (W. Ubachs).

Preprint submitted to Elsevier Science February 3, 2018 in particular of star-forming regions, where the photodissociation of CO is the governing dynamical process and CO is, at the same time, the coolant molecule [1]. The nitrogen molecule (N_2), iso-electronic to CO, has an onset of a dipole-allowed absorption spectrum at 100 nm, while its spectrum is similarly perturbed and prone to predissociation. The photoabsorption of N_2 is of great importance for the dynamics of the upper layers of the Earth’s atmosphere, and recently the importance of N_2 photo-predissociation for the interstellar medium is increasingly recognized [2]. The dipole-allowed absorption spectrum of molecular hydrogen (H_2) has its strongest lines in the Lyman and Werner bands in the wavelength range 90 − 115 nm. High-resolution spectroscopic studies and high-accuracy wavelength calibration studies of these sharp H_2 resonances has become of specific relevance for studies searching for a possible variation of the proton-electron mass ratio on a cosmological time scale [3, 4].

In the domain of high-resolution studies of atoms in particular the ground state of the Helium atom has been the target of many detailed studies over decades. This two-electron system is amenable for accurate ab initio calculations including quantum electrodynamic (QED) effects. The accurate determination of the ground state Lamb shift has been a driving force for the development of VUV-lasers and VUV-laser spectroscopic techniques since the first laser excitation of the first resonance line in the Helium atom [5].

2 Nano-second pulsed VUV sources

Techniques of harmonic generation in gases for the production of tunable radiation in the VUV domain have become well-established over the years. Although resonant sum- and difference-frequency techniques have many advantages to produce the best intensities and to cover a wide wavelength range [6], the approach of non-resonant third-harmonic generation has certain advantages for studies where absolute frequency calibration is the central concern. The setup displayed in Fig. 1 shows an example of a narrowband tunable laser system, based on a traveling-wave pulsed three-stage pulsed dye amplifier (PDA), which is seeded by the output of a continuous wave (CW) ring dyelaser [7]. By this means laser pulses of Fourier-transform limited bandwidth are produced, corresponding to the pulse duration (5 ns) of the frequency-doubled Nd:YAG laser employed for pumping the PDA. The intense PDA pulses (typically 200 mJ) are easily converted to the UV via second harmonic generation in a KDP crystal. Upon focusing the UV pulses underneath the orifice of a pulsed valve in a jet of xenon, third harmonics are generated, therewith producing VUV pulses at the exact 6^{th} harmonic of the fundamental. When running the ring dye laser in its operating range of λ_f = 560 − 690 nm, tunable radiation in the VUV range of λ_{V UV} = 93 − 115 nm can be produced. In view of the pulses being somewhat shortened in the non-linear upconversion process, the typical duration is ∼ 3 ns and the bandwidth in the VUV is about 250 MHz. This bandwidth slightly exceeds the Fourier-bandwidth due to chirp effects in the dye amplifier and harmonic upconversion.

Figure 1

The main advantage of the system displayed in Fig. 1 is that the VUV output is governed by the exact 6^{th} harmonic of the seed-frequency, a property which may be exploited in the accurate calibration of the measured resonances in the VUV. The frequency of the (CW) seed-light is accurately calibrated by performing saturated absorption spectroscopy on I_2, for which the individual hyperfine components are known to an accuracy of 1 MHz [9, 10]. The separation between an I_2 resonance at the the fundamental and the VUV-resonances at the 6^{th} harmonic is then bridged in terms of fringes from a stabilized etalon, locked to a stabilized HeNe-laser. This procedure is illustrated in Fig. 2, showing a recording of a specific line in the Lyman band system of the H_2 molecule. As in many experiments employing laser-based VUV radiation, a measurement scheme is chosen employing 1 VUV + 1 UV resonance-enhanced two-photon ionization. In this scheme the VUV radiation resonantly excites a high-lying state in the atomic/molecular system, which is then further ionized by the spatially and temporally overlapping UV beam, i.e. the leftover beam from the harmonic conversion in the gas jet. The error budget of VUV absolute frequency measurements contains contributions of statistics, and of systematic effects related to the AC Stark effect (induced primarily by the UV beam), and the residual Doppler effect due to non-perfect perpendicular alignment of the laser beams with respect to the collimated molecular beam. An additional source of uncertainty is related to possible chirp effects in the pulsed laser beams, as a result from time-dependent gain in the dye amplifiers. Experimental methods have been developed to determine the effect of this chirp on the measured transition frequency, and even to compensate the chirp effect by adapting the phase of the incident seed-light by an electro-optic modulator [8].

Figure 2

A comprehensive study has been performed on the VUV absorption spectrum of H_2 with the focus on accurate wavelength calibration for all the lines in the Lyman (B^1Σ^+_u - X^1Σ^+_g ) and Werner (C^1Π_u - X^1Σ^+_g ) bands [11]. Similar calibrations were performed on the HD isotopomer [12, 13]. These studies provide a database of accurate absolute wavelengths for all H_2 and HD absorption lines, at an accuracy of Δλ/λ = 4 × 10^{−8}. This database forms an ingredient for a study of possibly varying constants on a cosmological time scale. Thereby use is made of the equation: λ^z_i λ^0_i

= (1 + z_{abs})(1 + Δμ μ K_i) (1) making a comparison of the wavelengths λ^z_i as obtained from astronomical observations at high redshift z to the wavelengths λ^0_i in the database (at zero redshift z = 0), where z_{abs} is the overall redshift of an absorbing galaxy with observable H_2 abundance in the line of sight toward a quasar source. The final factor on the right hand side of the equation represents the possible effect of a drifting proton-electron mass ratio μ = m_p/m_e, with Δμ/μ the relative change in its value. For a comparison based on Eq. (1) a calculation of sensitivity coefficients K_i is required. Their values can be calculated from the known level structure of the H_2 molecule, by semi-empirical methods via [14]:

K_i = d ln λ_i d ln μ (2) Data on high-redshift H_2 absorptions are derived from astronomical observations using the largest optical dishes and high-resolution spectrometers (the Very Large Telescope with the UVES spectrometer, and the Keck telescope with the HIRES spectrometer). A comparison to accurate laboratory wavelengths, and invoking the calculated K_i sensitivity constants then leads to estimates on a possible variation of the proton-electron mass ratio. The most detailed studies [15–17] yield the overall result that the proton-electron mass ratio has not changed by more than 0.001 %, or |Δμ/μ| < 10^{−5} for redshifts of z = 2 − 3.5. This corresponds to look-back times of 10-12 billion years. This important result relies on the accurate laboratory calibration of H_2 and HD spectroscopic lines in the VUV.

Figure 3

Harmonic generation in the perturbative regime, hence at power densities of < 5 × 10^{12} W/cm^2, usually is restricted to 3^{rd} harmonics. However, with the high-power frequency-doubled PDA-laser system of Fig. 1 small amounts of 5^{th} harmonics could be produced, therewith producing the narrowest bandwidth pulses at wavelengths as short as 58 nm [18]. Based on such a VUV system the 1s^{2 1}S_0 - 1s2p^1P_1 resonance line of the helium atom could be excited at a resonance width of 600 MHz (see Fig. 3). Careful analysis of systematic effects, in particular the chirp-effects in the PDA-system and in the harmonic conversion process, allow for an experimental accuracy on the resonance line of Δλ/λ < 10^{−8} and a determination of the Lamb shift in the He ground state at an accuracy of 45 MHz [8].

3 Pico-second pulsed VUV sources

At incident power densities of > 10^{13} W/cm^2 a sequence of harmonics can be produced, known as the plateau region [19]. In order to reach the plateau threshold with laser pulses as narrow as possible in the frequency domain, a laser system was built generating high-power pulses at 300 ps duration [20]. A combination of techniques was employed as shown in the setup of Fig. 4. Powerful pump pulses at 300 ps duration were generated by stimulated Brillouin scattering in a compact configuration using liquid methanol as the nonlinear medium [21]. These pulses pumped a pulsed-dye amplifier, injection seeded by the output of a CW Ti:Sa laser, running on infrared dyes with typical excited state lifetimes of 300 ps. This allows for producing pulses of 300 ps, tunable in the range 700-850 nm at a repetition rate of 10 Hz. Further amplification in a Ti:Sa preamplifier and a seven-pass Ti:Sa bow-tie amplifier resulted in pulses of > 250 mJ/pulse at a Fourier-transform limited bandwidth of 1.5 GHz with a Fourier-product of Δν × Δτ = 0.45. After focusing the pulses with a lens with a focal distance of f = 20 cm, harmonics up to 15^{th} were generated [22]. In this process chirp phenomena cause a rather large broadening of the bandwidth of the pulses in the VUV [23]. This is the result of blue-shift effects from ionization in the early part of the pulse, and red-shift effects due to the plasma expansion in the trailing part of the pulses. Moreover, the combination of pulse duration and peak intensity causes the interaction volume to become fully ionized already during the front part of the pulse.

Figure 4

This VUV picosecond source was employed in a spectral recording of the He 1s^{2 1}S_0 - 1s4p^1P_1 resonance line at 52.2 nm, as shown in Fig. 3. The linewidth in the spectrum, amounting to 30 GHz, is mainly the result of the chirp effects in the harmonic generation process. This is a limiting factor to the achievable resolution for this design of a VUV source.

4 Femtosecond VUV pulses for high resolution spectroscopy

The ultra-short pulses from a femtosecond laser are seemingly not useful for spectroscopic studies in view of the Fourier-principle, dictating that such pulses exhibit an almost white spectrum. However, if the phase relationships between consecutive pulses are controlled and used in combination, the perspective alters drastically. This is ultimately reached in a frequency comb laser where full control is achieved for an infinite pulse train, thus delivering a mode spectrum: f_n = nf_{rep} + f_{CEO} (3) where f_{rep} is the repetition frequency of the modelocked frequency comb, f_{CEO} the carrier-envelope offset frequency, and n a large integer number. Both f_{rep} and f_{CEO} can be measured or locked against an atomic clock reference so that all modes of the comb laser are known with high precision [24, 25].

The production of high-order harmonics in the plateau region can be easily achieved for the ultrashort pulses from a frequency comb laser, in particular when a single pulse-pair is selected from the pulse train and subsequently amplified in a non-collinear optical parametric amplifier [26]. Under the condition that no extra phases are imposed upon the laser pulses in the amplification process nor in the harmonic conversion process, the resulting comb spectrum after conversion into the VUV is: f_m = mf_{rep} + qf_{CEO} (4) where m is again an integer and q is the integer harmonic order. The phase build-up in the amplifiers can be measured by interferometric techniques [26], while the phases acquired in harmonic conversion can be measured as well [27], and are therefore left out of consideration. It is noted however, that only the difference in phase build-up between consecutive pulses under consideration, Δψ, is of relevance.

In case two consecutive pulses are selected from a full pulse train emanating from a frequency comb laser the mode spectrum stays the same, except for the fact that the modes are no longer sharp spikes, but are diffused into a cosinemodulated mode spectrum. Upon harmonic conversion of the double-pulse structure the same mode structure as in Eq. (4) is retained [26] but again as a cosine modulation. This sequence of manipulative steps on the comb structure and the spectrum is highlighted in Fig. 5.

Figure 5

When using such double pulses for measuring a transition at frequency f_{tr} in an atomic system, one has to scan the time separation T between the two pulses so that an excitation spectrum is obtained of the form:

S(T ) ∝ cos[2πf_{tr}T ] (5)

where it is assumed that additional phase difference effects are controlled and may be neglected, so that Δψ(f_{tr}) = 0 [27]. The resulting cosine-modulated excitation spectrum is displayed in Fig. 6 representing a measurement of the 1s^{2 1}S_0 - 1s5p^1P_1 transition in helium, using the 15^{th} harmonic of a Ti:Sa frequency comb laser at a wavelength of 51 nm in the VUV.

Figure 6

This form of two-pulse delayed time-domain spectroscopy may be compared to Ramsey’s separated zone oscillatory field spectroscopy, where now the separation is in the time domain rather than in the spatial domain. In particular when using all pulses from a frequency comb laser it is usually referred to as direct frequency comb spectroscopy [28], and as in all absolute frequency measurements employing frequency comb lasers the central mode number, in this case the integer m of Eq. (4), must be determined. This is accomplished by performing the measurement for a number of repetition rates f_{rep} of the laser; for this purpose the cavity of the comb laser was redesigned to allow for settings between 100 MHz and 185 MHz [29]. This study results in an accurate value of the ionization potential of the helium atom as well as a determination of the He ground state (at the 10^{−9} accuracy level), including a value for the Lamb shift of the He ground state as accurate as 6 MHz [27, 29].

5 VUV Fourier-transform spectroscopy

From the perspective of non-laser based spectroscopies the technique of Fouriertransform (FT) spectroscopy has been been known as an important tool combining high resolving power and accurate intrinsic wavelength calibration. Moreover, the interferometric technique exhibits a multiplex advantage. The entire spectrum of the light source employed is covered at once in the spectral decomposition obtained by Fourier-transforming an interferogram. While FTspectroscopy is widely used in the infrared domain as an analytical sensing tool, it is used in the optical domain and even in the VUV domain down to wavelengths of 140 nm [30], but operation in all cases depends on a beamsplitter as a crucial optical element of the interferometer.

A recent development in VUV spectroscopy is the operation of a wave-front division setup to generate an interferogram between two spatial parts of a propagating beam at VUV wavelengths, without the use of transmitting materials. The scanning interferometer, shown in some detail in Fig. 7, exploits the delay imposed by two rooftop reflectors shifted with respect to each other. The stability and step-increments of the reflector displacement of the spatial scanning instrument should be kept under extreme control, i.e. within a fraction of the wavelength, in order to generate the interferograms. This is accomplished by probing the motion of one of the roof-shaped reflectors using a HeNe laser control interferometer. Such a FT-VUV device was developed at the Soleil synchrotron, where it is operated at the beam-line DESIRS at wavelengths in the range 40-200 nm [31].

Figure 7

The output of the undulator-based beam line delivers a beam of sufficient transverse spatial coherence to operate in wave-front division. A bell-shaped output spectrum of typically 7% bandwidth against the continuum background with sufficient power allows for the recording of interferograms on a VUV silicon photodiode, even in case of the lossy reflections on the roofshaped reflectors.

As an application example, a spectrum of high-resolution study of the HD molecule is shown in Fig. 8 [32]. An aspect of the FT-VUV spectrometer is that only linear absorption spectra can be recorded, imposing the disadvantage of Doppler broadening as a limitation. In the spectrum of Fig. 8, taken at room temperature, the Doppler width is the dominant factor in the linewidth amounting to 0.85 cm^{−1}. Recently, also static gas spectra with liquid- N_2 and liquid-He cooling of the windowless flow cell were recorded to reduce the linewidth. In addition the operation of the novel FT-VUV instrument was also demonstrated in a gas-jet configuration thereby obtaining a linewidth of 0.2 cm^{−1} in a spectrum of D_2 [33].

Figure 8

Acknowledgment

The authors wish to thank J. Bagdonaite, F. Brandi, P. C. Cacciani, G.D. Dickenson, C. Gohle, A. N. Heays, W. Hogervorst, U. Hollenstein, T. I. Ivanov, D. Joyeux, D. Z. Kandula, C. A. de Lange, M. T. Murphy, M. L. Niu, T. J. Pinkert, J. Philip, E. Reinhold, A. Renault, M. Roudjane, W. Vassen, M. O. Vieitez, S. Witte, A. L. Wolf, and R. T. Zinkstok for their involvement in the projects. Scientists and staff of the SOLEIL synchrotron are thanked for their support.

References

[1] R. Visser, E. F. van Dishoeck, and J. H. Black, Astron. Astrophys. 503, 323 (2009).

[2] X. Li, A. N. Heays, R. Visser, W. Ubachs, B. R. Lewis, S. T. Gibson, and E.

F. van Dishoeck, Astron. Astrophys. 555, A14 (2013). [3] W. Ubachs and E. Reinhold, Phys. Rev. Lett. 92, 101302 (2004). [4] E. Reinhold, R. Buning, U. Hollenstein, P. Petitjean, A. Ivanchik, and W.

Ubachs, Phys. Rev. Lett. 96, 151101 (2006).

[5] K. S. E. Eikema, W. Ubachs, W. Vassen, and W. Hogervorst, Phys. Rev. Lett.

71, 1690 (1993).

[6] U. Hollenstein, H. Palm, and F. Merkt, Rev. Scient. Instr. 71, 4023 (2000). [7] W. Ubachs, K. S. E. Eikema, W. Hogervorst, and P. C. Cacciani, J. Opt. Soc.

Am. B 14, 2469 (1997).

[8] K. S. E. Eikema, W. Ubachs, W. Vassen, and W. Hogervorst, Phys. Rev. A 55, 1866 (1997).

[9] I. Velchev, R. van Dierendonck, W. Hogervorst, and W. Ubachs, J. Mol. Spectrosc. 187, 21 (1998).

[10] S. C. Xu, R. van Dierendonck, W. Hogervorst, and W. Ubachs, J. Mol. Spectrosc. 201, 256 (2000).

[11] J. Philip, J. P. Sprengers, Th. Pielage, C. A. de Lange, W. Ubachs, and E.

Reinhold, Can. J. Chem. 82, 713 (2004).

[12] U. Hollenstein, E. Reinhold, C. A. de Lange, and W. Ubachs, J. Phys. B 39, L195 (2006).

[13] T. I. Ivanov, M. Roudjane, M. O. Vieitez, C. A. de Lange, W. U. L. Tchang- Brillet, and W. Ubachs, Phys. Rev. Lett. 100, 093007 (2008).

[14] W. Ubachs, R. Buning, K. S. E. Eikema, and E. Reinhold, J. Mol. Spectrosc.

241, 155 (2007).

[15] J. A. King, J. K. Webb, M. T. Murphy, and R. F. Carswell, Phys. Rev. Lett.

101, 251304 (2008).

[16] A. L. Malec, R. Buning, M. T. Murphy, N. Milutinovic, S. L. Ellison, J. X.

Prochaska, L. Kaper, J. Tumlinson, R. F. Carswell, and W. Ubachs, Monthly Not. Royal Astron. Society, 403, 1541 (2010).

[17] F. van Weerdenburg, M. T. Murphy, A. L. Malec, L. Kaper, and W. Ubachs, Phys. Rev. Lett. 106, 180802 (2011).

[18] K. S. E. Eikema, W. Ubachs, W. Vassen, and W. Hogervorst, Phys. Rev. Lett.

76, 1216 (1996).

[19] A. L’Huillier, L. A. Lompré, G. Mainfray, and C. Manus, J. Phys. B 16, 1363 (1983).

[20] F. Brandi, I. Velchev, D. Neshev, W. Hogervorst, and W. Ubachs, Rev. Scient.

Instr. 74, 32 (2003).

[21] D. Neshev, I. Velchev, W. Majewski, W. Hogervorst, and W. Ubachs, Appl.

Phys. B68, 671 (1999).

[22] F. Brandi, D. Neshev, and W. Ubachs, Phys. Rev. Lett. 91, 163901 (2003). [23] F. Brandi, F. Giammanco, and W. Ubachs, Phys. Rev. Lett. 96, 123904 (2006). [24] T. W. Hänsch, Rev. Mod. Phys. 78, 1297 (2006). [25] J. H. Hall, Rev. Mod. Phys. 78, 1279 (2006).

[26] D. Z. Kandula, A. Renault, C. Gohle, A. L. Wolf, S. Witte, W. Hogervorst, W. Ubachs, and K. S. E. Eikema, Opt. Expr. 16, 7071 (2008).

[27] D. Z. Kandula, C. Gohle, T. J. Pinkert, W. Ubachs, and K. S. E. Eikema, Phys. Rev. A 84, 062512 (2011).

[28] A. Marian, M. C. Stowe, J. R. Lawall, D. Felinto, and J. Ye, Science 306, 2063 (2004).

[29] D. Z. Kandula, C. Gohle, T. J. Pinkert, W. Ubachs, and K. S. E. Eikema, Phys. Rev. Lett. 105, 063001 (2010).

[30] A. Thorne, J. Anal. At. Spectrum. 13, 407 (1998).

[31] N. de Oliveira, M. Roudjane, D. Joyeux, D. Phalippou, J.-C. Rodier, and L.

Nahon, Nat. Phot. 5, 149 (2011).

[32] T. I. Ivanov, G. D. Dickenson, M. Roudjane, N. de Oliveira, D. Joyeux, L.

Nahon, W. U. L. Tchang-Brillet, and W. Ubachs, Mol. Phys. 108, 771 (2010). [33] G. Dickenson, T. I. Ivanov, W. Ubachs, M. Roudjane, N. de Oliveira, D. Joyeux, L. Nahon, W. U. L. Tchang-Brillet, M. Glass-Maujean, H. Schmoranzer, A. Knie, S. Kubler, and A. Ehresmann, Mol. Phys. 109, 2693 (2011).

Figure 1. Experimental setup of the pulsed dye-amplifier based narrowband and tunable VUV source, in a crossed beam configuration for performing on-line Doppler reduced spectroscopy. In the upper part the calibration facilities, a HeNe stabilized etalon and an I_2 saturation spectroscopic setup are shown. The box in the lower right corner shows the differentially pumped vacuum setup for the production of VUV in the first zone, and the excitation of a perpendicularly crossing molecular beam and the ion extraction, time-of-flight and detection system. The acousto-optic modulator (AOM) serves to shift the carrier frequency in the chirp detection scheme. The electro-optic modulator serves to modulate the phase of the seed-light in order to compensate for chirp effects in the amplifier [8].

Figure 2. Measurement of the B-X (6,0) P(1) line of H_2, at 94 472.029 cm^{−1}, recorded by 1 VUV + 1 UV resonance enhanced photoionization. The absolute calibration is performed against the I_2 line marked with an asterisk; this is the a_1 hyperfine component of the P(34) line in the B-X (10,3) band at 16 245.13532 cm^{−1} [10].

Figure 3. Excitation resonances of he atomic lines. (a) 1 VUV + 1 UV photoionization of the 1s^{2 1}S_0 - 1s2p^1P_1 line of helium at 54.8 nm produced with the ns-PDA laser system upon 5^{th} harmonic conversion at a linewidth of 600 MHz; (b) Absorption spectrum of the 1s^{2 1}S_0 - 1s4p^1P_1 line of helium at 52.2 nm, recorded with the picosecond VUV laser system as shown in Fig. 4, at a linewidth of 30 GHz.

Figure 4. Two-column figure. Setup of the VUV source for pulses in the 300 ps time domain. A pulsed dye amplifier (PDA) is injection seeded by the output of a CW titanium-sapphire (Ti:Sa) laser at λ = 760 − 830 nm and pumped by the SBS compressed output of a frequency-doubled Nd:YAG laser. These pulses are further amplified in a bowtie amplifier with Ti:Sa crystals. The infrared laser pulses produce, upon focusing at f = 20 cm, power densities of > 10^{13} W/cm^2, sufficient to reach the plateau for generating high harmonics into the wavelength window of λ = 50 nm. Figure reproduced from Ref. [22].

Figure 5. Sequence of manipulation of the comb structure from a frequency comb, with the full pulse train emanating from the comb laser, the amplification of a pair of two pulses in a non-collinear optical parametric amplifier, and the conversion into harmonics in the plateau region. The 15^{th} harmonic, to be used in the excitation of the helium atom, exhibits a cosine-modulated mode spectrum with period f_{rep}.

Figure 6. Two-pulse excitation probability on the 1s^{2 1}S_0 − 1s5p^1P_1 transition in Helium. The lower x-axis represents the scan of the repetition rate of the frequency comb laser, while on the upper axis the delay between the two pulses is shown in attoseconds. Figure from Ref. [27].

Figure 7. Two-column figure. Experimental layout of the wavefront-division interferometer used at the DESIRS beam line at Soleil with the roof-shaped VUV reflecting surfaces in the path to produce the interferences at the detector (PD). In the zoomed-in part (upper right corner) the control of the beampath difference, established by a HeNe laser beam reflecting multiple times from the rear of the traveling optic, is shown. Figure from Ref. [32].

Figure 8. Fourier-transform VUV absorption spectrum of the HD molecule at room temperature measured at the DESIRS beamline at Soleil. The top panel displays the bell-shaped curve associated with a single setting of the undulator. The bottom spectrum is a zoom-in displaying a line in the Lyman band (L15R2) and a line in the Werner band (W3Q2) at the Doppler width of 0.85 cm^{−1}.